The switch in the upper right select between 6 microseconds (us) for fast displays (Tektronix scopes) and 20 us for slow displays (VR14). The switch in the upper left selects between negative and positive intensity pulses. The connector on the left side of the board is for the +-15 VDC high quality (low noise, stable output) power supply.
Feel free to contact me, David Gesswein firstname.lastname@example.org with any questions, comments on the web site, or if you have related equipment, documentation, software etc. you are willing to part with. I am interested in anything PDP-8 related, computers, peripherals used with them, DEC or third party, or documentation.
PDP-8 Home Page  PDP-8 Site Map  PDP-8 Site Search